Prolong your HPLC column’s lifetime and improve data quality by making simple changes to your HPLC sample preparation procedures
Jun 14 2017 Read 1042 Times
HPLC columns are central equipment in many laboratories, protecting them from damage is a prime consideration. Extending column lifetime can save time and costs in a busy laboratory. A well-maintained column also produces consistent and accurate data, supporting confident analyses.
How can I prolong column lifetime?
- Sample preparation
Prefiltering samples prior to injection onto the column can help to limit the addition of particulates to the system. A syringe filter or syringeless filter vials suitable for this step, depending on the sample type, number of samples, and sample injection method.
Centrifuging the sample with or without filtration also helps to reduce the number of particulates. Particles that might otherwise make it into the system and onto the column collect at the bottom of the centrifuge vial.
- Mobile phase
The mobile phase is another factor that influences column lifetime.
Filtering and degassing the mobile phase will remove particulates and air, ensuring they do not reach and damage the column. Filtration is especially important if the mobile phase contains buffer salts or modifiers from dissolved solid material.
Also, it can be useful to consider the compatibility of a sample with the mobile phase, because some sample components are reactive with particular HPLC solvents.
- In-line filters and guard columns
In-line filters and guard columns help prevent particulates entering the system and the column. The use of these column protectors depends on the application, with either one, or both, required.
A guard column is a short column that the sample and mobile phase pass through before entering the main column, helping to protect against damage by adsorbing particulates. The use of a guard column depends on the specific application, because it can be an expensive protection option requiring consideration of the costs against benefits.
Maintaining a clean column is another way to extend a column’s lifetime. Occasionally flushing the column with mobile phases that differ from those used on the system (e.g., more water or organics) can help to clean the column. Extensive flushing in a backwards direction can also help with cleaning, especially if you notice increasing column back pressure.
Storing columns in an appropriate solvent can help prolong their life. If a column is stored for long intervals (e.g., over two days), consult the column’s handling instructions to identify the best storage solvent for that particular column.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Articles Chromatography - Visiting the 15th International Symposium - New Application Book Offers Expertise in GPC Analysis - Same Separation, Speedier Solution Mass S...
View all digital editions
Mar 22 2018 Istanbul, Turkey
Mar 23 2018 Beijing, China
Mar 27 2018 Guangzhou, China
Mar 28 2018 Lyon, France
Apr 02 2018 Suntec City, Singapore