Pattern Approval for Measuring Protein in Wheat and Barley
Jun 20 2006
The Cropscan 2000B NIR Analyser measures protein and moisture in whole grains, including wheat, barley, soybean, corn and sorghum. The NMI Pattern Approval certifies that the Cropscan 2000B is suitable for use as an instrument to measure protein in wheat and barley for trading purposes.
The Pattern Approval process requires that the instrument be able to make a measurement of protein in wheat with a Maximum Permissible Error (MPE) of 0.4% and in barley with a MPE of 0.5%. The conditions under which these measurements are tested include:
- Instrument temperature between 10C and 40C,
- Humidity 85%RH exposure for 2 days
- Mains Voltage: -15% to +10% variation in mains voltage
- Electrical Burst
- Electromagnetic Fields
- Electrostatic Discharge
The NMI M15-1 Pattern Approval is the first part of a certification process. The instrument needs to be tested for accuracy under the NMI V10 Uniform Test Procedure for the Verification, Certification and In-Service Inspection of Protein Measuring Instrument for Grain. This is to be done by a licensed inspector or certifier using a set of Trade Measurement Certified Reference Materials.
NIR Technology Australia provides the Cropscan 2000B NIR Analyser with calibrations for measuring protein and moisture in wheat and barley. The NMI M15-1 Pattern Approval is provided for all Cropscan 2000B instruments. These calibrations are tested against Trade Measuring Certified Reference Materials to comply with the NMI V10 procedure prior to shipment.
Lab Asia 30.3 - June 2023
In This Edition Chromatography Articles - Refining Process Separation Procedures with Prepacked Chromatography Columns Mass Spectrometry & Spectroscopy Articles - Analysis of Acetaldehy...
View all digital editions
Jun 14 2023 Bangkok, Thailand
Jun 18 2023 Dusseldorf, Germany
Jun 19 2023 Shanghai, China
Jul 04 2023 Bethesda, MA, USA
Jul 04 2023 Cardiff, UK