Specular Reflection Spectroscopy with Viewing
Mar 11 2019 Read 543 Times
Printed ink on a coated aluminum sample was analysed using a specular reflection accessory with the unique capability of simultaneous sample viewing to confidently position the sample. Visual images and spectra of the sample were recorded and stored.
FTIR specular reflection sampling represents an important technique useful for measurement of films on reflective substrates and bulk solid materials. The optical design of commercial specular reflection accessories requires the reflected infrared beam to be collected at an angle that is the same as the angle of incidence. In most cases, the sample is placed face down on a sampling mask or optical body of the accessory making it difficult to determine the exact sampling point. The aim of this application note is to collect an infrared spectrum of various printed colors on a coated aluminum surface using a specular reflection accessory with viewing capabilities for precise sample positioning.
A coated aluminum sample containing multiple colors printed on the surface was analysed using the PIKE Technologies’ GladiATR Vision configured with a specular reflection viewing plate (45 degree angle of incidence). The viewing camera is situated underneath the 3 mm sampling area of the specular reflection plate. The camera magnification is 110X and the viewing area is 770 x 590 μm. The sample is illuminated through fibers using LED lights. Data collection range was 4000–400 cm-1 at a spectral resolution of 4 cm-1. Background and sample collection time was 30 s.
Figure 1. GladiATR Vision accessory from PIKE Technologies
The GladiATR Vision is a novel sampling tool which couples small area infrared analysis with simultaneous viewing. It may be fitted with a diamond or germanium ATR crystal or a specular reflection plate. With an optical configuration similar to a beam condenser, the GladiATR efficiently demagnifies the beam by more than 3X for small spot-size focusing, about 3 mm, allowing collection of high quality spectra. Its all-reflective optical design enables sampling across the full spectral range of the near-IR, mid-IR and far-IR.
This application required various colors on a printed aluminum panel to be analysed. Due to the close proximity of the colors on the sample, the GladiATR Vision as a sampling accessory was advantageous over traditional specular reflection accessories because it allowed the desired sample area to be pinpointed and documented using a camera focused onto the sample. Figure 2 shows the spectra of green printed ink and an unprinted area. From the difference between these two spectra, the unique features of the green printed ink may be observed in the CN stretch region, specifically the strong absorbance band located at 2100 cm-1. The inset shows a typical image that was viewed as the sampling position was located.
*Figure 2. Spectra of green printed ink (red) and an unprinted area (blue) of a coated aluminum panel; inset is a typical image from the GladiATR Vision.
The GladiATR Vision offers the unique capability of sample viewing coupled with measuring in either specular reflection or ATR mode.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In this Issue Articles Mass Spectrometry & Spectroscopy 67th ASMS Conference on Mass Spectrometry and Allied Topics Chromatography Using Gas Chromatography for measuring atmo...
View all digital editions
Aug 25 2019 San Diego, CA, USA
Sep 01 2019 Berlin, Germany
Sep 03 2019 Manchester, UK
Sep 04 2019 Chiba City, Japan
Sep 08 2019 Krakow, Poland