Mass Spectrometry & Spectroscopy
Copolymer Identification by ATR/FT-IR Spectroscopy
Jul 03 2019 Read 409 Times
Polymers are ubiquitous in the food, medical, and scientific industries. The ability to identify different polymers for QC/QA is a critical step required in many applications. Attenuated Total Reflection (ATR) is a popular sampling method for FT-IR analysis used to chemically identify or distinguish between polymer samples. Here we use the IRIS monolithic diamond ATR to analyse solid cellulose and cellulose propionate copolymer samples.
Polymer samples were obtained from Scientific Polymer Products Inc. Within the Polymer Sample Kit, pure cellulose and cellulose propionate copolymer samples correspond to Cat# 661 and Cat# 321, respectively. ATR background spectra were collected without any sample present. Pure cellulose and cellulose propionate copolymer transmittance spectra were collected under the same conditions using the IRIS monolithic diamond ATR accessory (Figure 1). All spectra were collected on a commercial FTIR spectrometer. Scans were acquired over 30 seconds at a resolution of 4 cm-1.
The primary vibrational bands of pure cellulose are highlighted in Figure 2 by the vertical gray-lines. These bands are associated with the free –OH stretch (3335 cm-1), C–H stretch (2900 cm-1), and C–O primary-alcohol stretch (1025 cm-1).1 These vibrational bands are also present in cellulose propionate along with three large bands originating from the propionate substituent
marked by the vertical red-lines shown in Figure 2. The two higher frequency propionate bands are associated with the C–H stretch (2940 cm-1), C=O carbonyl stretch (1745 cm-1). At lower frequencies, the presence of a polarization-dependent set of bands show up corresponding to the perpendicular and parallel oriented C–C axial stretch occurring at 884 cm-1 and 806 cm-1, respectively.2
The IRIS monolithic diamond is an excellent ATR accessory for a wide range of samples including powders, gels, liquids, solids, and more. Specifically, IRIS’ QA/QC abilities have been shown by distinguishing cellulose from cellulose propionate copolymer. Spectral bands were identified for each polymer.
To learn more about The IRIS Diamond ATR, click here.
Visit PIKE Technologies
1. Garside, P., Wyeth, P. Studies in Conservation, 48(4), 269-275 (2003)
2. Nobukawa, S., et al. Polymer Journal, 47(4), 294-301 (2015)
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Mass Spectrometry & Spectroscopy - 67th ASMS Conference on Mass Spectrometry and Allied Topics - Modular MS Bench System with Integrated Gas Supply for Sciex LC-MS/MS Unveile...
View all digital editions
Aug 04 2019 Anaheim, CA, USA
Aug 04 2019 Portland, OR, USA
Aug 05 2019 Lombard, IL, USA
Aug 25 2019 San Diego, CA, USA
Sep 01 2019 Berlin, Germany