Taking ATR in a new directon: Specac’s Arrow™
Jul 14 2020 Read 362 Times
The Specac Arrow™ is a revolutionary new consumable ATR solution for the Quest™ accessory with applications ranging from testing of chemically aggressive samples through to large scale batch testing of liquid samples and solids dried from solution. Arrow prevents sample cross-contamination and removes the need to clean between samples, ensuring high accuracy of results and speeding up analysis.
Arrow consists of a thin silicon ATR element fixed into a slide made from 100% recycled polypropylene. This allows for mounting on the Quest™ ATR unit and also for convenient handling and storage without risking contact with the sample or the silicon material.
As an example of how Arrow can be used, we have demonstrated the quantification of water/ethanol mixtures. Accurate quantification of the composition of food and drink is essential for compliance with local regulations on labelling of goods for general sale and for correct application of customs duties when imported. Food and drink labelling should provide correct information about calorific and nutritional content to help people stay within recommended limits.
Solutions of water/ethanol in 26 different concentrations were prepared. A batch of Arrow™ slides were filled with aliquots from each solution and then covered with the Arrow™ sample cap to prevent evaporation. One slide was retained for use as a background. The prepared batch was then transported to the spectrometer for analysis.
Spectra were recorded on a commercially available spectrometer. The background spectrum was recorded first, followed by each of the 26 Arrow™ sample slides in turn. The complete procedure takes less than 10 minutes because there is no wipe down and cleaning of the sampling point is required between each spectral acquisition.
Figure 1 shows selected spectra for different concentrations of Ethanol solution. The pure water spectrum (black line) shows characteristic peaks at ca. 3350 cm-1 (assigned to two overlapped peaks: symmetric, ν1, and antisymmetric, ν3 stretching vibrations) and ca.1640 cm-1 (assigned to the ν2 bending vibration). The pure spectrum of ethanol (purple line) displays peaks due to an OH stretching vibration at ca. 3330 cm-1, CH stretching vibrations typical of an alkyl group between 3000-2800 cm-1 and further peaks in the fingerprint region below 1500 cm-1. The spectra of the mixture samples are a combination of the two pure samples, however due to interactions between the two some small changes are observed.
A 2-Factor PLS model was constructed from the spectra (Figure 2), explaining 99.9% of the variance within the dataset. The model was validated using fresh samples (Figure 3), and an excellent correlation was observed with an R2 = 0.997, and a RMSEP = 4.71 x 10-1.
This demonstrates that Arrow™ can be used to successfully quantify alcohol concentrations in a simple model system using PLS and identify all relevant spectral bands. For applications requiring the bulk testing and spectral discrimination of more complex systems such adulterated or contaminated foodstuffs, the speed of spectral acquisition and convenience of batch sample preparation come into their own.
Do you like or dislike what you have read? Why not post a comment to tell others / the manufacturer and our Editor what you think. To leave comments please complete the form below. Providing the content is approved, your comment will be on screen in less than 24 hours. Leaving comments on product information and articles can assist with future editorial and article content. Post questions, thoughts or simply whether you like the content.
In This Edition Chromatography - Measurement of Pesticides in Cannabis sativa and Hemp Matrices - Biphenyl Modified Silica Stabilised by Bulky Substituents - Wettable Phase for Analytical a...
View all digital editions
Aug 11 2020 Guanghzou, China
Aug 13 2020 Shanghai, China
Aug 16 2020 San Francisco, CA, USA
Aug 19 2020 Mumbai, India
Aug 19 2020 Mumbai, India