Mass Spectrometry & Spectroscopy
How to Overcome LC-MS/MS Matrix Effects to Achieve Maximum Reliability
Aug 22 2018 Read 1020 Times
Common approaches to counter matrix effects include matrix-matched calibration, standard addition to each sample and the application of internal standards. The first two strategies entail additional work in terms of sample preparation or additional runs, incurring further costs. Importantly, matrix-matched calibration and standard addition cannot fully compensate for matrix effects. The effectiveness of internal standards in compensating for matrix effects depends on the choice of calibrants.
Internal standards can be chemically related compounds such as derivatives (e.g. zearalanone for zearalenone) or similar compounds with identical behaviour during the ionisation process that differ only in terms of the mass of the atoms (stable isotopes). LC-MS/MS analyses rely on stable isotope dilution assays to overcome matrix effects by the addition of known amounts of stable isotope-labelled standards to the analysed sample. Stable isotope labelling involves the use of non-radioactive isotopes like ²H, 13C or 15N to replace the naturally occurring atom. Using Deuterium (²H) to replace the naturally occurring 1H doubles the mass, which is the reason why deuterated labelled standards might show retention time shifts resulting in less accurate LC-MS/MS results.
Carbon, on the other hand, is naturally present in most compounds and mycotoxins. Replacing naturally-occurring 12C by 13C changes the total mass of the atom only slightly, unlike deuterated labelled standards. Added 13C-labelled standards, e.g. 13C-labelled mycotoxins, retain the same characteristics as their 12C analogues, eluting at the same retention time from the separation column.
The difference in mass between 12C and 13C mycotoxins, as shown for deoxynivalenol in the image, allows the separation and identification of both eluted forms when the detection is performed with mass spectrometry.
The 13C peak represents the known quantity of the added standard. This peak can then be used to calculate the unknown amount of the analyte and thereby compensate for different ionisation efficiencies. In this way, innovative 13C standards can eliminate matrix effects.
Visit the Romer Labs website to learn more about 13C-labelled standards.
Romer Labs Singapore Pte Ltd was referenced in the following research papers
In This Edition Articles - Expanding the Boundaries of Light Scattering for Macromolecules - Electrostatic Charges and Their Effects on Weighing - How the world’s most advanced analytical...
View all digital editions
Feb 28 2021 Virtual event
Mar 03 2021 Guanghzou, China
Mar 08 2021 Virtual Event
Mar 22 2021 Digital event
Apr 14 2021 Tokyo, Japan